THE GENERATION DIAGNOSTIC SYSTEM
The i-chamber is an auxiliary device for the ichroma (ichroma ™, ichroma ™ II) systems and provides an environment set to the correct temperature to the iChroma cartridge when heating is required between tests. You can improve the reliability of your test results by keeping the temperature constant, one of the environmental factors that can cause unexpected errors in your test results.
|Dimensions||190 mm (L) x 120.6 mm (W) x 108 mm (H)|
|Weight||1.5 kg (± 0.3 kg)|
|Number of cartridge slots||6 (Temperature can be set independently for each slot)|
|Power supply||DC 12 V, 5 A|
|Data output||LCD (16 x 4 characters)|
|Communication ports||Mini USB port (for software upgrade)|
|Environmental conditions for the operation||Temperature and humidity : 15~35°C(59~95oF), Humidity up to 70%,
Installation : Dry, clean, flat and horizontal surface in an indoor setting away from direct sunlight, mechanical vibration and electromagnetic field, etc.
- About 10 minutes, when the target temperature (Td) is within 5°C of the ambient temperature
- About 1 hour, if the target temperature (Td) is lower than the ambient temperature by 10°C or more
Yes. Press the two buttons simultaneously to set the target temperature, then press the two buttons again to complete the temperature setting.
No. please turn on the i-Chamber and wait until the target temperature is reached. Test should be carried out when the i-Chamber provides a constant temperature condition.
Yes. Press the right button for 3 seconds, set the reaction time, and press the two buttons simultaneously to complete the time setting.